A device that is placed in the wall that separates the trachea and esophagus in order to enable a total laryngectomy patient to make voice. The tracheoesophageal voice prosthesis (TEP) uses a one-way valve to let air pushed up from the lungs to pass through from the trachea and enter the esophagus, causing the walls of the esophagus to vibrate as a new voice, but without letting food or liquids to pass through the other way, from the esophagus to the trachea.
The need for a tracheoesophageal voice prosthesis:
A patient who undergoes a total laryngectomy procedure will have the entire larynx removed and the end of the trachea redirected to an opening—called a stoma—created in the front of the neck. The result is that the patient can now breathe through this stoma, but would be unable to make any voice, because the larynx (voice box) is gone, and the air for voicing that once passed up through the larynx, causing the vocal cords to vibrate and thereby making voice, now simply exits the trachea at the stoma. Such a patient can obtain an alternate voice by using a TEP device.
The placement of a tracheoesophageal voice prosthesis:
In order to place a TEP device into the wall that separates the trachea and esophagus (the tracheoesophageal party wall), a puncture must first be made, into which the device can fit. This puncture can be made during the laryngectomy procedure (primary tracheoesophageal puncture) or else afterward in a separate procedure (secondary tracheoesophageal puncture), typically some weeks after the laryngectomy. Some clinicians prefer to place the device itself as part of the primary puncture. Others first place a catheter to hold the puncture open and allow for tube feedings while the pharynx heals, and then they place the actual device in an office setting 10 days or more later.
The use of a tracheoesophageal voice prosthesis:
When a patient with a TEP device wants to speak, he or she must momentarily cover the stoma in the front of the neck, so that air coming up from the lungs is redirected through the TEP and into the esophagus to make voice. Some patients will simply cover the stoma as needed with their thumb, if their stoma is normally left entirely open, or is only concealed from view with a small, unobtrusive, breathable cloth. More commonly, however, the patient’s stoma is fitted with a housing, into which is placed a heat and moisture exchanger (HME); in these cases, the HME, which looks like a flat, plastic button, can be pushed as needed to block the stoma and enable voicing. Whatever the case, speaking with a TEP device is a learned skill and requires training through speech therapy.
Tracheoesophageal voice prosthesis (1 of 4)
Tracheal flange beginning to embed (at arrows) because the TEP device is too short.
Tracheoesophageal voice prosthesis (2 of 4)
Esophageal flange is bowing outward because the TEP device is too short.
Tracheoesophageal voice prosthesis, corrected fitting (3 of 4)
Same patient, with normal (now flat) esophageal flange fitting.
Hypopharynx of a tracheoesophageal voice prosthesis patient (1 of 4)
Panoramic view of the hypopharynx, in a patient who has undergone total laryngectomy. The entrance to the esophagus is at the line of arrows.
Hypopharynx, as the tracheoesophageal voice prosthesis patient makes voice (2 of 4)
Air has been diverted from the trachea and through the tracheoesophageal voice prosthesis (shown in the next two images) so that the hypopharyngeal tissue here is now vibrating (thus, is blurred) and making voice.
Tracheoesophageal voice prosthesis (3 of 4)
Seen here is the inner (esophageal) flange of the tracheoesophageal prosthesis, with its central flutter valve in closed (swallowing or resting) position.
Tracheoesophageal voice prosthesis, as the patient makes voice (4 of 4)
The patient has capped his tracheostome and is diverting air into the esophagus through the now-open central flutter valve. The pharyngoesophageal tissues are now vibrating (as seen in photo 2), and this vibration is blurring the image.
TEP buried (1 of 4)
This man’s TEP voice is becoming gradually more effortful, choppy, and strained. The explanation is that tissue (mucosa) is slowly burying the inner (esophageal) flange of the device. Half of the circumference is hidden under the mucosal flap indicated by dotted line.
Re-inserted and re-positioned TEP (2 of 4)
After re-inserting and re-positioning, now the entire circumference is visible, and voice has returned to effortless baseline.
Tissue overgrowing flange (3 of 4)
In a different patient, tissue (at X’s) is also trying to overgrow the internal (esophageal) flange.